

ORIGINAL ARTICLE 

Year : 2018  Volume
: 41
 Issue : 2  Page : 7783 


Assessment of the performance characteristics of an optically stimulated luminescence dosimetry system in Gabon
Philippe Ondo Meye^{1}, Cyril Schandorf^{2}, Roger Ondo Ndong^{3}, Hans Essone Obame^{3}
^{1} General Directorate of Radiation Protection and Nuclear Safety, Ministry of Water and Energy, Libreville, Gabon ^{2} Department of Medical Physics, School of Nuclear and Allied Sciences, Accra, Ghana ^{3} Département Sciences Physiques, Ecole Normale Supérieure, Libreville, Gabon
Date of Submission  05Mar2018 
Date of Decision  06Apr2018 
Date of Acceptance  07May2018 
Date of Web Publication  24Aug2018 
Correspondence Address: Mr. Philippe Ondo Meye General Directorate of Radiation Protection and Nuclear Safety, Ministry of Water and Energy, BP 1172, Libreville Gabon
Source of Support: None, Conflict of Interest: None  Check 
DOI: 10.4103/rpe.RPE_28_18
To ensure the reliability of the performance of the optically stimulated luminescence (OSL) dosimetry system used for individual monitoring of occupationally exposed workers in Gabon, some key performance indicators were evaluated. The performance indicators assessed included: measurement uncertainty, linearity, coefficient of variation, and photon energy and angular dependence. The results for uncertainty assessment were in very good agreement with the International Commission on Radiological Protection (ICRP) and International Atomic Energy Agency (IAEA) requirements for overall accuracy. The maximum relative overall uncertainty in absolute value in the region of the recording level was 45%. In the region beyond the recording level, the maximum relative overall uncertainty was 42% (in absolute value). The results for linearity assessment met the requirement for linearity of International Electrotechnical Commission (IEC) 62387 standard. The results for coefficient of variation assessment were within the relevant requirement of IEC 62387 for doses beyond the recording level. The results obtained for photon energy and angle of incidence were in good agreement with the IAEA and IEC 62387 corresponding requirements. The OSL dosimetry system can be relied on to provide acceptable monitoring data for the individual monitoring service in Gabon. Keywords: Coefficient of variation, energy and angular dependence, intercomparison, linearity, measurement uncertainty
How to cite this article: Meye PO, Schandorf C, Ndong RO, Obame HE. Assessment of the performance characteristics of an optically stimulated luminescence dosimetry system in Gabon. Radiat Prot Environ 2018;41:7783 
How to cite this URL: Meye PO, Schandorf C, Ndong RO, Obame HE. Assessment of the performance characteristics of an optically stimulated luminescence dosimetry system in Gabon. Radiat Prot Environ [serial online] 2018 [cited 2022 May 18];41:7783. Available from: https://www.rpe.org.in/text.asp?2018/41/2/77/239683 
Introduction   
Between 2013 and 2016, the Individual Monitoring Service (IMS) of the General Directorate of Radiation Protection and Nuclear Safety (DGRSN), Ministry of Water and Energy in Gabon, participated in four regional intercomparison exercises on measurements of the personal dose equivalent H_{p}(10) in photon fields. Two of them were organized in 2013 and 2016 by the International Atomic Energy Agency (IAEA) through the regional technical cooperation projects RAF/9/043,^{[1]} Strengthening the Transfer of Experience Related to Occupational Radiation Protection of the Nuclear Industry and other Applications Involving Ionizing Radiation, and RAF/9/053, Strengthening Technical Capabilities for Patients and Occupational Radiation Protection in Member States. The first exercise in 2013 involved 24 countries from Africa and three countries from outside this continent. Seventeen African countries participated in the second exercise. One intercomparison exercise was organized in 2013 by Landauer Europe ^{[2]} to assess the performance of some of the African laboratories that use Landauer OSL dosimetry systems. Five laboratories from five African countries participated in this exercise. The last intercomparison exercise was organized in 2016 by IMSs of the DGRSN and that of the Radiation Protection Institute of the Ghana Atomic Energy Commission. Only Gabon and Ghana participated in this exercise. The purpose of this work was to highlight the results obtained by Gabon during these intercomparison exercises and to assess some performance indicators using data from these intercomparisons and relevant international standards for personal monitoring. The work was limited to data from IAEA and Landauer Europe intercomparisons. The intercomparison organized by national IMSs in Gabon and Ghana is discussed in reference.^{[3]}
Materials and Methods   
Materials
The dosimetry system considered here is an InLight dosimetry system (Landauer, Inc., USA). It is designed to assess personnel exposure to beta particles and photons using OSL.^{[4],[5],[6]} The system consists of individual passive dosemeters, a reader, a highintensity light annealer, and a software installed on a personal computer.
Reader
The reader under consideration is the portable InLight Microstar (32.7 cm width × 23.2 cm depth × 10.9 cm height,^{[4],[6]} serial number 11040681, date of installation, April 2012). The OSL readout of each detector is performed using an array of 38 green LEDs (532 nm wavelength) operated in continuous waveOSL mode during about 1 s. All the 38 LEDs are used for the stimulation of low doses (strong beam), whereas only 6 LEDs are used for high doses (weak beam). The dose range is evaluated before the measurement using one LED and the cutoff point is an adjustable parameter.
Dosemeters
The dosemeter considered comprises a case and a slide. The latter contains four detector elements of Al_{2}O_{3}: C cut into round pieces of ~5 mm in diameter sandwiched between two layers of polyester for a total thickness of 0.3 mm. The detectors are located in read positions 1 (E1), 2 (E2), 3 (E3), and 4 (E4). When the slide is inside the case, each detector is positioned behind different filters providing different radiation attenuation conditions. The signal from each OSL detector is used in conjunction with a dose algorithm to evaluate different dosimetric quantities (H_{p}(10), H_{p}(0.07), and H_{p}(3)).^{[6]} This dose algorithm inherently uses individual calibration factors (ICFs). For wider and homogeneous radiation fields, the chest OSLD badge (element E3) may also act as a surrogate for eye lens doses.
Methods
Details on Landauer and the 2013 and 2016 International Atomic Energy Agency intercomparison exercises
During the 2013 and 2016, IAEA regional intercomparison on measurements of the personal dose equivalent H_{p}(10) in photon fields, the dosemeters were irradiated at the surface of slab phantoms using values of dose rates calculated from the measured air kerma determined with the Secondary Standard Dosimetry Laboratory (SSDL) reference instrument. For the 2013 exercise,^{[1]} irradiations were performed by the SSDL of the Nuclear Research Centre of Algiers as follows:
 Irradiations using the ^{137} Cs source to dose values varying from 0.5 to 10 mSv in normal incidence (linearity verification –dosemeters irradiated in group of 3)
 Irradiations at the fixed dose of 2 mSv using three Xray qualities (N60, N80, and N150) (energy response – dosemeters irradiated in group of 3)
 Irradiations with SCs quality at the fixed dose of 2 mSv and angular incidences of 0°, 45°, and 60° (angular response – dosemeters irradiated in group of 2) and
 Irradiations at different doses (0.7–150 mSv), energies (N150, SCs, and SCo), and in mixed qualities (N150 + SCs) (Blind test – dosemeters irradiated in group of 2).
For the 2016 exercise, irradiations were carried out as follows:
 Irradiations using the ^{137} Cs source to dose values lying between 0.2 and 10 mSv in normal incidence (linearity verification – dosemeters irradiated in group of 4)
 Irradiations at the fixed dose of 2 mSv using three Xray qualities (N60, N80, and N120) (energy response – dosemeters irradiated in group of 4)
 Irradiations with N60 and SCs qualities at the fixed dose of 2 mSv and angular incidences of 0°, 45°, and 60° (angular response – dosemeters irradiated in group of 2); and
 Irradiations with SCo quality and mixed qualities (N60 + SCs) at different doses (20 and 5 mSv, respectively) (Blind test – dosemeters irradiated in group of 4).
Landauer Europe intercomparison was limited to the assessment of dosimetry systems involved using a linearity response test.^{[2]} Irradiations were performed by Landauer Europe using a ^{137} Cs source in normal incidence. The doses delivered were 0.19 mSv, 6.27 mSv, and 19.90 mSv (dosemeters irradiated in group of 3).
Uncertainty assessment
The analytical method described in International Electrotechnical Commission (IEC) technical report TR 62461 was followed.^{[7]} Three steps were required: The construction of a model function, the collection of data and existing knowledge, and the calculation of uncertainty associated with the result of the measurement. The model function used in this study was derived from that given in the standard IEC 62387:^{[9]}
H_{p} (10) = N_{0} K_{n}K_{E,}φ K_{env} f(H¯_{m})(1)
Where H_{p}(10) is the measuring quantity personal dose equivalent, N_{0} is the reference calibration factor, K_{n} is the correction factor for nonlinearity, K_{E,}φ is the correction factor for photon energy and angle of incidence, K_{env} is the correction factor for environmental conditions, H¯_{m} is the mean indicated value, mean reading of the dosemeter in units of H_{p}(10), f(H¯_{m}) is a function derived from the calibration curve. This model function is similar to that presented in reference.^{[3]} The notable difference is that the correction factor for photon energy and angle of incidence in Equation (1) is a function of energy. Therefore, an “if” algorithm with four branches was used.^{[8]} The main data and required information needed in the present work for the uncertainty analysis were provided by the standard IEC 62387^{[9]} (K_{env}), the IEC technical report TR 62461 (N_{0}), the three intercomparison exercises considered here (K_{n} and K_{E,}φ), and IAEA safety guide No. RSG1.3.^{[10]} In particular, IEC TR 62461 suggests limits of ±5% with a triangular distribution for N_{0}, and IAEA RSG1.3 assumes a rectangular distribution for type B uncertainties. These assumptions were used in this study. As H¯_{m} is a mean value of several measurements, its standard deviation is the standard deviation of a single measurement divided by the square root of the number of measurements. This standard deviation of the mean is then taken as the standard uncertainty associated with H¯_{m}. The standard uncertainty σ_{f} associated with f(H¯_{m}) is then derived using the uncertainty propagation formula
Where σ_{H¯m} is the standard uncertainty associated with H¯_{m}. f(H¯_{m}) depends on the calibration curve. For the purpose of this work, f(H¯_{m}) was chosen so that
f(H¯_{m}) is the same for Landauer and 2013 IAEA intercomparison exercises as they occur in the same period. To calculate the uncertainty associated with H_{p}(10), first sensitivity coefficients of input quantities were computed using the formula
Where X_{i}= {N_{0}, K_{n}, K_{E,}φ, K_{env}} denote the true values (capital letters) and x_{i}= {n_{0}, k_{n}, k_{E,}φ, k_{env}} their respective expectation values (small letters). c_{xi} is the sensitivity coefficient associated with x_{i}. Thereafter, the uncertainty contribution to the output quantity H_{p}(10) was calculated as follows:
Where s_{xi} is the standard uncertainty associated with x_{i}. The total standard uncertainty u associated with h_{p}(10) was obtained by taking the geometrical sum of these contributions:
Finally, the complete result of the measurement is given by
Where k_{cov} is a coverage factor. A comparison of the results with International Commission on Radiological Protection (ICRP) and IAEA ^{[10],[11],[12]} requirements for overall accuracy was performed. These are:
 relative uncertainty limits of –33% and +50% (a factor of 1.5 in either direction) in the region near the relevant dose limit
 relative uncertainty limits of ±100% (factors of 2 and 0) in the region of the recording level.
These requirements are met by the trumpet curve given by
Where H_{t} is the conventional true value (or delivered dose provided by an SSDL) and H_{0} (=0.05 mSv for the dosimetry system considered) is the recording level.
Data used were collected from intercomparison exercises results for linearity response (quality: SCs; dose rage: 0.19–19.9 mSv), energy and angular response (qualities: N60, N80, N120, N150 and SCs; angles: 0, 30, 45, and 60°; dose: 2 mSv), and blind test (qualities: N150, N150+ SCs, SCs, SCo, N60+ SCs; angles: 0° and 30°; doses: 0.7, 2.5, 5, and 20 mSv).
Linearity and coefficient of variation
Linearity
The procedure from IEC 62387 was followed.^{[9]} The dosemeters are irradiated at known dose equivalents and the variation of the response due to the change of the dose equivalent shall not exceed the values given in the standard. This requirement is met only if the relationship below is valid,
Where
 H_{t, i} is the conventional true value of the personal dose equivalent delivered to dosemeters of group i
 H_{t, ref } is the reference conventional true value of dose equivalent
 H¯_{m, i} is the mean of the indicated values of dosemeters of group i
 H¯_{m, ref} is the mean of indicated values of dosemeters of the reference group
 U_{com} is the expanded uncertainty of H¯_{m, i}/H¯_{m, ref}
 U_{t, com} is the combined relative expanded uncertainty of H_{t, ref}/H_{t, i}
The corresponding limits in terms of mean response were derived (0.88 and 1.15). U_{com} was computed following the procedure given in Annex A.
Coefficient of variation
According to the IEC 62387 standard, this test shall be performed together with the linearity test. The statistical fluctuations of the indicated value shall fulfill the following requirement:
Photon energy and angle of incidence
IEC 62387 requirement was followed:^{[9]} the variation of relative response due to a change of radiation energy and angle of incidence within the rated ranges shall not exceed the value given for H_{p}(10) in the standard. The radiation qualities N60, N80, N100, N150, SCs (^{137} Cs), and SCo (^{60} Co) are part of the qualities used during the 2013 and 2016 IAEA intercomparisons. The above requirement is met if, for every radiation quality, the inequality is valid,
Where r_{min} (0.69 and 0.71 for 48 keV and energies ≥65 keV, respectively) and r_{max} (1.82 and 1.67 for 48 keV and energies ≥65 keV, respectively) are, respectively, the minimum and maximum relative response given in the standard. The other quantities are as defined above. From equation (11), limits in terms of mean response were derived. The IAEA ^{[10]} requirements for radiation energy and angle of incidence were also tested in this work. Requirements on energy and directional dependence are combined and given by equation
Where
R¯_{E,}φ is the mean response at photon energy E and angle of incidence φ between 0 and 60° by step of 20°. For energy E, all ISO series can be used.
[Table 1] summarizes the recommended ranges of all the parameters that have been investigated in the study.  Table 1: Summary of the recommended ranges of all the parameters that have been assessed in the study
Click here to view 
Results and Discussion   
Uncertainty assessment
[Table 2] shows uncertainty budget for the reference measured dose of 2013 intercomparison exercise. From all the uncertainty budgets of each measured dose, it is observed that the maximum relative overall uncertainty in absolute value in the region of the recording level is 45%. In the region beyond the recording level, the maximum relative overall uncertainty is 42% (in absolute value). These results are in very good agreement with ICRP and IAEA requirements for overall accuracy. These results are also close to that obtained in reference ^{[3]} where only one energy (662 keV) at normal incidence was used. [Figure 1] and [Figure 2] show the results in terms of response for the 2013 IAEA exercise. In particular, it can be seen that more accurate values of dose were obtained when using correction factors. This trend was also observed for the 2016 IAEA and Landauer exercises (not shown here). In addition, for each intercomparison exercise considered here, 100 % of the data points were within the trumpet curve.  Table 2: Uncertainty budget of the reference measured dose of the 2013 International Atomic Energy Agency intercomparison exercise. A relative overall uncertainty of ±33.85% ([0.713/2.107]×100) was obtained
Click here to view 
 Figure 1: Results of uncertainty assessment in terms of response for the 2013 International Atomic Energy Agency intercomparison exercise. The results are presented for doses varying from 0.5 mSv to 10 mSv, for the radiation qualities N60, N80, N150, SCs, SCo and N150+ SCs, and for radiation angle of incidence lying between 0 and 60°. Comparison with the trumpet curve is also presented. One hundred percent of the data points, that is, 34 are within the trumpet curve
Click here to view 
 Figure 2: Results of uncertainty assessment in terms of response for the 2013 International Atomic Energy Agency intercomparison exercise using correction factors. The results are presented for doses varying from 0.5 mSv to 10 mSv, for the radiation qualities N60, N80, N150, SCs, SCo and N150+ SCs, and for radiation angle of incidence lying between 0 and 60°. Comparison with the trumpet curve is also presented. One hundred percent of the data points, that is, 34 are within the trumpet curve. This case shows more accurate values than the previous one
Click here to view 
Linearity and coefficient of variation
[Figure 3] presents the results obtained for linearity. It is observed that they are in very good agreement with IEC 62387 requirements for linearity. Data points and their associated statistical uncertainty near the recording level could be significantly improved by increasing the number of dosemeters in the irradiation process when delivering a given dose. Indeed, statistical or type A uncertainty can, in principle, be reduced by increasing the number of measurements. Furthermore, Annex A of IEC 62387 was used to compute statistical uncertainty. In this approach, the statistical uncertainty is obtained by multiplying the standard deviation of a specific group of measurements by the ratio of a coverage factor by the square root of the number of measurements. The coverage factor is obtained from the Student's tdistribution. For a number of 2, 3, and 4 measurements, this ratio is 8.98, 2.48 and 1.59, respectively, while for a number of 6 and 10 measurements, the ratio is 1.05 and 0.715, respectively. Furthermore, the results obtained in this study are comparable to that obtained in reference.^{[3]} [Figure 4] presents the results for the coefficient of variation according to the IEC 62387 standard. All the data points are within the required limits apart from two data points near the recording level. Again, it is believed that the results could be improved if more dosemeters were used during the irradiation process for each delivered dose.  Figure 4: Results for coefficient of variation using International Electrotechnical Commission 62387 standard and data collected from International Atomic Energy Agency and Landauer intercomparison exercises. The results are presented for doses varying from 0.19 mSv to 19.9 mSv. Irradiations were carried out using a ^{137}Cs source in normal incidence
Click here to view 
Photon energy and angle of incidence
[Table 3] and [Table 4] present the results obtained according to the IEC 62387 standard. All the data points are within the required limits. However, the statistical overall uncertainty limits associated with mean responses at (662 keV, 45°) and (48keV, 45°) for the 2013 and 2016 intercomparisons, respectively, are clearly outside the required limits. Mean responses at 662 keV, 60° and 118 keV, 30°, for the 2013 IAEA exercise, and 662 keV, 45° for the 2016 IAEA exercise have their lower statistical overall uncertainty limit outside the required limits by a small margin. Increasing the number of dosemeters will help improve these results. [Table 5] presents the results obtained according to the IAEA (safety guide No. RSG1.3) requirements for radiation energy and angle of incidence. The results show that the requirement is met since all the data points lie within the required limits.  Table 3: Results for photon energy and angle of incidence according to the IEC 62387 standard using data from 2013 International Atomic Energy Agency intercomparison exercise. The required limits are 0.69 and 1.97 and 0.71 and 1.80 for 48 keV and energies ≥65 keV, respectively
Click here to view 
 Table 4: Results for photon energy and angle of incidence according to the IEC 62387 standard using data from 2016 International Atomic Energy Agency intercomparison exercise. The required limits are 0.65 and 1.85 and 0.67 and 1.70 for 48 keV and energies ≥65 keV, respectively
Click here to view 
 Table 5: Results for photon energy and angle of incidence according to the International Atomic Energy Agency requirement using data from 2013 and 2016 International Atomic Energy Agency intercomparison exercises. The required limits are 0.69 and 1.31
Click here to view 
Conclusions   
The OSL dosimetry system used for the national IMS in Gabon performed satisfactorily with respect to the performance indicators assessed, namely, measurement uncertainty, linearity, coefficient of variation, photon energy and angular dependence. The results obtained can be improved if appropriate correction factors are applied. The results for uncertainty assessment were in very good agreement with ICRP and IAEA requirements for overall accuracy. The results for linearity were in good agreement with the requirement for linearity specified in the IEC 62387 standard. The results for coefficient of variation met the relevant requirement specified in the IEC 62387 standard, except for doses near the recording level. The results obtained for photon energy and angle of incidence were in good agreement with the IAEA and IEC 62387 corresponding requirements. It is believed that all the results obtained could be improved if the number n of measurements (n ≥ 6) was increased.
Finally, the OSL dosimetry system can be relied on to provide acceptable monitoring data for the IMS in Gabon. To ensure the longterm performance and reliability of the OSL dosimetry system, QC checks recommended by the manufacturer must be followed in between regional and international intercomparisons.
Financial support and sponsorship
Nil.
Conflicts of interest
There are no conflicts of interest.
References   
1.  Arib M, Herrati A, Dari F, Ma J, LounisMokrani Z. Intercomparison 2013 on measurements of the personal dose equivalent Hp(10) in photon fields in the African region. Radiat Prot Dosimetry 2015;163:27683. 
2.  Landauer Europe. Intercomparaison des Laboratoires de Suivi Dosimétrique du Continent Africain. Rapport Ref: REPSCI007. Landauer Europe; 2013. 
3.  Ondo Meye P, Schandorf C, Amoako JK, Manteaw PO, Amoatey EA, Adjei DN, et al. Intercomparison on the measurement of the quantity personal dose equivalent Hp(10) in photon fields. Linearity dependence, lower limit of detection and uncertainty in measurement of dosimetry systems of individual monitoring services in Gabon and Ghana. Radiat Prot Dosimetry 2017;177:25060. 
4.  Landauer. In Light ^{®} Systems – MicroStar ^{®} User Manual. Landauer, Inc; 2009. 
5.  Landauer, Inc. In Light Model 2 Dosemeter, Characterization and Uncertainty Analysis for National Dosimetry Services (NDS), Health Canada. Rev. 004. Landauer, Inc; 2009. 
6.  Yukihara EG, McKeever SW. Optically Stimulated Luminescence – Fundamentals and Applications. Chichester, UK: Wiley; 2011. 
7.  International Electrotechnical Commission. Radiation Protection Instrumentation – Determination of Uncertainty in Measurement. Technical Report TR 62461. Geneva, Switzerland: IEC; 2015. 
8.  Ondo Meye P, Schandorf C, Ondo Ndong R. Reader quality control tests, dose algorithm comparison, and signal depletion of optically stimulated luminescence dosimetry systems used for individual monitoring: A case of the dosimetry system of the national individual monitoring service in Gabon. Radiat Prot Environ 2018;41:8893. 
9.  International Electrotechnical Commission. Radiation Protection Instrumentation – Passive Integrating Dosimetry Systems for Personal and Environmental Monitoring of Photon and Beta Radiation. International Standard IEC 62387. Geneva, Switzerland: IEC; 2012. 
10.  International Atomic Energy Agency. Assessment of Occupational Exposure Due to External Sources of Radiation. IAEA Safety Standards Series. Safety Guide No. RSG1.3. Vienna, Austria: IAEA; 1999. 
11.  International Atomic Energy Agency. Occupational Radiation Protection. Draft Safety Guide DS453. Vienna, Austria: IAEA; 2014. 
12.  International Commission on Radiological Protection. General Principles for the Radiation Protection of Workers. ICRP publication 75. Ann. ICRP 27;1997. 
[Figure 1], [Figure 2], [Figure 3], [Figure 4]
[Table 1], [Table 2], [Table 3], [Table 4], [Table 5]
